8210 system is suitable for voltage, charge, IEPE and TEDS types vibration transducer calibration. Adopt back-to-back comparison method (ISO 16063-21:2003) to complete acceleration transducer (or velocity transducer or vibration meter) calibration, to fast get accurate sensitivity, amplitude linearity and frequency response calibration for accelerometers and velocity sensors.
Vibration Transducer Calibration System 8210
▪Working piezoelectric acceleration transducer, velocity transducer and vibrometer while produced, maintained and employed
▪Various types of acceleration, velocity transducer and vibrometer, such as voltage type, charge type, IEPE type, TEDS type and so on.
▪ISO 16063-21:2003 Vibration and Shock Transducer Calibration Method
▪Calibrate accurately all kinds of acceleration transducer, velocity transducer and vibrometer, including voltage type, charge type, IEPE, TEDS, etc.
▪Frequency range is 2Hz-20kHz, support point-by-point comparison calibration, swept sine calibration, random excitation and impact excitation methods, etc;
▪High precision and dynamic range, system uncertainty: <1% (160Hz, 100 m/s2)
▪At the end of calibration, it will automatically generate a detailed report, including sensitivity, linearity and amplitude and frequency response curve; all data will be input database, compiling refers to ISO17025.
▪Substitution calibration option prolongs the service life of standard transducer.
▪TEDS transducer support, and results wrote in it.
▪Sensitivity Calibration: frequency range default160Hz, acceleration default 10m/s2
▫Swept Sine: Breakpoint tabular form defined the sweep range and reference acceleration value; User defined sweep method (logarithmic or linear), sweep speed and time.
▫Step Sine: Tabular form defined at least 10 frequency points and its reference acceleration, max.4096 discrete points, frequency points according to 1/3 octave or 1/1 octave.
▫Random Calibration（FFT）：Breakpoint tabular form defined random profile; User defined RMS value, analysis lines and calibration time.
▪Amplitude Linearity: The curve is frequency VS acceleration which reflects the frequency VS sensitivity. Tabular form defined the measurement frequency points.
▪Transverse Sensitivity Calibration: The transverse sensitivity is the accelerometer output sensitivity when acceleration is applied at its geometrical right angle.
Frequency Range (Hz)
2 to 20k
Matched Power Amplifier
Input to Blower(VA)
Overcurrent Transistor temperature Overload
Maximum Power Consumption(VA)
12.5 or app.
Temperature range: 0℃ to 40 ℃ Humidity range: 20 to 85%RH without dewdrop
Copyright © DynaTronic Corporation (China) Ltd. All Rights Reserved.